The IMU / YAW unit can be easily configured to run in ‘standalone’ CAN operating modes using the supplied RLVBCAB30-S cable and VBOX Setup software.
Configuring an IMU / YAW with Racelogic Module Setup Software
- Connect the unit to a PC using the supplied RLVBCAB030-S cable.
- Power up the unit using a suitable 12 V power source.
- Run VBOX Setup software.
- Select the correct COM port the unit is assigned to (check device manger if unsure) and the software will display the current settings for the attached unit. The screenshot shows the settings for an IMU04.
- Make the changes required and then click the “write to unit” button.
User Polled Mode
When selecting Timed Mode, the option to set the Timer value will appear. The timer value is in milliseconds (ms). A smaller value means data will be sent more frequently, a larger value means data will be sent less frequently. The range of values that can be entered is 0 – 30000, however the minimum value that should be entered is 10. Any figure below this may cause data to be repeated on successive cycles.
If a value of 0 is entered this will be changed to 1 on the next power cycle.
Frequency output can be calculated as follows:
Freq = (1/Timer) * 1000
The timer value for a required frequency can be calculated as follows:
Timer = (1/Freq) * 1000
Some example timer values are shown against the frequency output below:
|Timer Value (ms)||Frequency (Hz)|
Note: When any change is made using VBOX Setup Software the IMU / YAW unit must be powered cycled for the changes to take effect.